私たちの目的は、皆様方にデバイスCitizen Blood Pressure Monitor CH-618の取扱説明書に含まれたコンテンツを可能な限り早く提供することです。オンラインプレビューを使用すると、Citizen Blood Pressure Monitor CH-618に関してあなたが抱えている問題に対する解決策の内容が素早く表示されます。
説明書Citizen Blood Pressure Monitor CH-618をこちらのサイトで閲覧するのに不都合がある場合は、2つの解決策があります：
多くの人々は画面表示ではなく印刷された説明書を読むほうを好みます。説明書を印刷するオプションも提供されており、上記のリンクをクリックすることによりそれを利用できます - 説明書を印刷。説明書Citizen Blood Pressure Monitor CH-618を全部印刷する必要はなく、選択したページだけを印刷できます。紙を節約しましょう。
次のページにある説明書Citizen Blood Pressure Monitor CH-618の内容のプレビューは、以下にあります。次のページにある説明書の内容をすぐに表示したい場合は、こちらをご利用ください。
Contents ................ 01....... Blood Pressure Monitor Intended Use 02....... 1. What is blood pressure ? 02....... 2. Why is it useful to measure blood pressure at home ? .......... A. WHO blood pressure classi cations .......... B. Variations in blood pressure 04....... 3. Important information before use of the unit 06....... 4. Device description .......... A. Name of the parts .......... B. Description of display symbols 08....... 5
Contents ................ 10....... 8. Symbol Di nitions. 12....... 9. How to measure. .......... A. Inserting batteries in position. .......... B. Setting date and time. .......... C. Steps to take blood pressure measurement. .......... D. Storing recalling and erasing measurement data. .......... E. To change batteries. 15....... 10. Maintenance. 15....... 11. Safe-keeping. 15....... 12. To prevent malfunction. 16....... 13.
MEDICAL DISCLAIMER This manual and product are not meant to be a substitute for advice provided by your doctor or other medical professional. You are not to use the information contained herein, or this product for diagnosing or treating a health problem or prescribing any medication. If you have or suspect that you have a medical problem, promptly consult your healthcare provider. Blood Pressure Monitor Intended Use: Measures human's Systolic blood pressure, Diastolic blood pressure and
1. What is blood pressure ? Blood pressure is a measurement of the force of blood owing against the walls of the arteries. Arterial blood pressure is constantly changing during the course of the cardiac cycle. The highest pressure in the cycle is called the systolic blood pressure, the lowest is the diastolic blood pressure. Both pressure readings, the systolic and the diastolic are necessary to enable a physician to evaluate the status of a patient's blood pressure. Many factors such as
as near as possible to measuring the blood pressure in an environment that is close to this, is why it is useful to take the measurement at home. A. WHO blood pressure classiﬁcations Standards for assessment of high or low blood pressure without regard to age, have been established by the World Health Organization (WHO), as shown in the chart. * Figure No.1 B. Variations in blood pressure Individual blood pressures vary greatly both on a daily and a seasonal basis. These variations are
The thick line represents sleep. The rise in blood pressure at 4 PM (A in the graph) and 12 PM (B in the graph) correspond to an attack of pain. 3. Important information before use of the unit Please rest for at least 5~10 minutes before taking the measurement. To allow your blood vessels to return to the condition prior to taking the measurement, please wait at least 3~5 minutes between measurements. You may need to adjust the wait time according to your personal physiological situation.
Perform measurement quietly in a relaxed position. Do not wind the cuff over jacket or sweater sleeve, or measurement cannot be done. Keep in mind, that blood pressure naturally varies from time to time throughout the day and also is affected by lots of different factors such as smoking, alcohol consumption, medication and physical activity. People with a condition that causes circulatory problems (diabetes, kidney disease, arteriosclerosis or poor peripheral circulation) may get lower rea
/min DIA Pulse mmHg : M STOP mmHg START AVG PM SYS 1 2 USER 4. Device description: A. Name of the parts LCD DISPLAY FUNCTION BUTTON RECALL M MEMORY BUTTON SETTING BUTTON START/STOP BUTTON ( Stand-by) WRIST CUFF CARRYING CASE ALKALINE BATTERY UM-4 (AAA) 1.5 V x 2 * Figure No.3 6
B. Description of display symbols 1 USER 1 SYSTOLIC RATE 2 USER 2 LAST 3 READING AVG AVERAGE DIASTOLIC LOW BATTERY RATE SYMBOL HOUR MINUTE : PM (ALSO REPRESENTS AS YEAR HEART RATE UNDER SETTING MODE) MONTH/DATE HEART RATE SYMBOL 7
5. Changing batteries Remove the battery cover at the left side of the unit and insert batteries into the battery compartment in accordance with ribbon strip as shown, taking extreme care that the polarities + and - are observed. Important notice: to take the batteries out easily, just pull the ribbon strip for help. * Figure No.4 6. Attaching pressure cuff A. Wrap the pressure cuff around the wrist. a. The display of the unit should be placed on the palm side of the wrist.
B. Fasten the pressure cuff snugly. a. Do not pull strongly on the pressure cuff. 13.5 ~ 19.5 cm (5.3 ~ 7.7 inch) b. Do not make the pressure cuff too tight. * Figure No.5-1 . Correct measurement position 7 It is extremely important for the measured values, that the cuff is at the same height of the heart. Higher or deeper position will give deviations from the true value. Measurement while sitting down: a. Place your elbow on a table or other object (such as our carry case).
8. Symbol Deﬁnitions SYMBOLS Deﬁnitions This symbol appears when the battery power is excessively low or the polarity reverses. Low Battery We suggest you replace all batteries with Symbol new ones, and make sure the +/- polarities are properly positioned. Once pulse is detected, the symbol ashes with each pulse beat. Our suggestion: Heart Rate Please do not talk or move during Symbol measurements. This symbol appears when LCD displays AVG average value of last 3 readings. Last 3 Rea
SYMBOLS/ CONDITIONS/ INDICATION/ SYMPTOMS CAUSES CORRECTION Cuff has been placed Wrap the cuff properly incorrectly. so that it is positioned correctly. Measuring Error Symbol shown Did you talk or move Measure again. Keep on display during measurement? wrist steady during or the blood measurement. pressure value Shaking of the wrist is displayed with the cuff on. excessively low or high. Air circuit abnormality. Measure again. Measuring Error Symbol In ation pressure Switch the u
9. How to measure A. Inserting batteries in position: a. Open battery cover. b. Refer to the picture in positioning batteries. (Figure No.4) c. All LCD segments appear on the display in 3 seconds. B. Setting Date and Time: a. Press " " button over 2 secs ("year" starts ashing) Press " " button to set the correct year (2011,2012,…2026) b. Press " " button ("month" starts ashing) Press " " button to set the correct month (1,2,3,.....12) c. Press " " button a
START c. Press" "button. Press " " button to select USER STOP START 2 ( 1 ) . Press " " button again. All elements STOP display '888' in 3 seconds. Afterward, displays the updated time. The measuring blood pressure symbol will ash on the display and then the air pressure will automatically pump up to 195 mmHg. If blood pressure tends to be on the high level, the air pressure will automatically increase to the required level for measurement. * D
b. Recalling data: (1). Press " M " button.If there is no data in the memory, only TIME/DATE and USER symbol will appear on the display. If yes, the rst reading will be the average of the latest 3 measurements. (The average value dose not appear if there are two or fewer measurements stored in memory.) (2). Press " M " button again, the latest measurement will appear with the number as well as the recording date and time. The data called up on the display is numbered accordingly
10.MAINTENANCE: Use a piece of cloth with water or mild cleansing agent to clean the case and then use a piece of dry cloth to wipe it dry. Do not use gas or any strong cleansers in cleaning. When the unit is not to be used for a long time, remove the batteries. (Leaking of battery liquid can cause trouble) 11.SAFE-KEEPING: Always keep the unit in the carrying case after its use. Do not put the item directly under the sunlight, in high temperature, or humid and dusty places. Do not store in e
13.WARNING: The device measures the blood pressure of adult only. The device is not supposed to be used if your wrist has any wound or injury. In the event of the cuff pumping up and not stopping, please open the cuff at once. To ensure precise measurement reading, recalibration of the device is recommended after 2 years from the date of purchase. Shipping plus handling cost and recalibration service fee shall be charged accordingly. Limited Warranty This device is warranted to be free from
|1||Citizen Blood Pressure Monitor CH-403C||Blood Pressure Monitor||270|
|2||Citizen Blood Pressure Monitor CH-406D||Blood Pressure Monitor||12|
|3||Citizen Blood Pressure Monitor CH-437C||Blood Pressure Monitor||24|
|4||Citizen Blood Pressure Monitor CH-432B||Blood Pressure Monitor||190|
|5||Citizen Blood Pressure Monitor CH-452||Blood Pressure Monitor||237|