Our goal is to provide you with a quick access to the content of the user manual for Agilent Technologies Water Dispenser 8596E. Using the online preview, you can quickly view the contents and go to the page where you will find the solution to your problem with Agilent Technologies Water Dispenser 8596E.
If looking through the Agilent Technologies Water Dispenser 8596E user manual directly on this website is not convenient for you, there are two possible solutions:
Many people prefer to read the documents not on the screen, but in the printed version. The option to print the manual has also been provided, and you can use it by clicking the link above - Print the manual. You do not have to print the entire manual Agilent Technologies Water Dispenser 8596E but the selected pages only. paper.
Below you will find previews of the content of the user manuals presented on the following pages to Agilent Technologies Water Dispenser 8596E. If you want to quickly view the content of pages found on the following pages of the manual, you can use them.
Agilent 8590 E-Series
Portable Spectrum Analyzers
These specifications apply to the Agilent 8594E
dc coupled 9 kHz to 2.9 GHz
Technologies 8591E, 8593E, 8594E, 8595E, and
ac coupled 100 kHz to 2.9 GHz
8596E spectrum analyzers.
dc coupled 9 kHz to 6.5 GHz
ac coupled 100 kHz to 6.5 GHz
All specifications apply over O°C to +55°C. The
analyzer will meet its specifications after 2 hours
dc coupled 9 kHz to 12.8 GHz
of storage at a constant temperature, within the
Accuracy ±20% Frequency Readout Selectivity (Characteristic) Accuracy –60 dB/–3 dB (Start, Stop, Center, Marker) ±(frequency readout x frequency 1 3 kHz to 10 kHz 15:1 reference error +span accuracy +1% 100 kHz to 3 MHz 15.1 of span +20% of RBW+100 Hz x N*) 1 kHz, 30 kHz 16:1 –40 dB/–3 dB Marker Count Accuracy 30 Hz to 300 Hz 10:1 Frequency Span ≤10 MHz x N* ±(marker frequency x frequency 1 Video Bandwidth Range 30 Hz to 1 MHz in 1,3 sequence reference error + counter resolution 1 Hz to 1 MHz (O
19.1 TO 22 GHz 12.4 TO 19.4 GHz 6.0 TO 12.8 GHz 400 kHz TO 6.5 GHz Displayed Average Noise Level Amplitude Specifications (Input terminated, 0 dB attenuation, 1 Hz/30 Hz VBW, sample- Amplitude specifications do not apply for Analog+ detector) mode and negative peak detector mode except as 8591E 30 Hz RBW 1 kHz RBW noted in “Amplitude Characteristics.” 400 kHz to 1 MHz ≤–130 dBm ≤–115 dBm 1 MHz to 1.5 GHz ≤–130 dBm ≤–115 dBm Amplitude Range 1.5 GHz to 1.8 GHz ≤–128 dBm ≤–113 dBm Displayed averag
Spurious Responses Reference Level Second Harmonic Distortion Range same as amplitude range 5 MHz to 1.8 GHz (91E) <–70 dBc for –45 dBm tone at Resolution 0.1 dB for log scale, 0.12% of 2 input mixer. reference level for linear scale 10 MHz to 2.9 GHz (93E) <–70 dBc for –40 dBm tone at Accuracy ±0.3 dB @ –20 dBm 2 >10 MHz (94E, 95E, 96E) input mixer. 0 dBm to –59.9 dBm ±(0.3 dB +.01 x dB from –20 dBm) >2.75 GHz (93E, 95E,96E) <–100 dBc for –10 dBm tone at 2 input mixer (or below displayed Fr
Dynamic Range (Characteristic) Option Specifications 5 Dynamic Range TG Feedthrough Option 010 and 011 Tracking Generator 8591E 106 dB ≤–106 dBm Frequency Range 8591E (Opt. 011) 100 dB ≤–57.24 dBmV 8591E 100 kHz to 1.8 GHz 8593E (> 400 kHz) 111 dB ≤–112 dBm (Opt. 011, 75 Ω ) 1 MHz to 1.8 GHz 8594E (> 400 kHz) 106 dB ≤–107 dBm 8593E, 94E, 95E, 96E 9 kHz to 2.9 GHz (> 5 MHz) 111 dB ≤–112 dBm 8595E (>400 kHz) 109 dB ≤–110 dBm Output Level 8596E (> 400 kHz) 109 dB ≤–110 dBm Range 8591E 0 to –70 dBm
Power Requirements Keyboard (Opt. 041 or 043) 5 Pin mini-DIN, compatible with ON (Line 1) 90 to 132 V rms, 47 to 440 Hz HP C1405B and most IBM/AT key 195 to 250 V rms, 47 to 66 Hz boards Power consumption <500 VA; Ext. Trigger Input BNC, TTL levels, positive edge <180W trigger Standby (Line 0) Power consumption <7 W GPIB and Parallel (Opt 041) SH1, AH1, T6, L4, ST1, RL1, PPO, DC1, C1 C2, C3, & C28 and 25 Pin User Program Memory 238 Kbytes non-volatile RAM subminature D-shell female for paralle
Agilent Technologies’ Test and Measurement Related Literature Pub. Number Support, Services, and Assistance General Purpose Information Agilent Technologies aims to maximize the value you receive, Agilent 8590L and 8592L while minimizing your risk and problems. We strive to ensure that you get the test and measurement capabilities you paid Product Overview 5962-7275E for and obtain the support you need. Our extensive support Agilent 8590C/E/L and EM Series resources and services can help you
|1||Agilent Technologies Water Dispenser 8156A||Water Dispenser||12|
|2||Agilent Technologies Water Dispenser 8591E||Water Dispenser||23|
|3||Agilent Technologies Water Dispenser 8594E||Water Dispenser||12|
|4||Agilent Technologies Water Dispenser 8593E||Water Dispenser||38|
|5||Agilent Technologies Water Dispenser 856290216||Water Dispenser||34|